เครื่องวัดปริมาณรังสี DKG-21M Personal Gamma Radiation Dosimeter
Сonforms to the CE standards
Meets the requirements of IEC 61526 standard
ТУ У 33.2-22362867-010:2007
Direct reading personal dosimeter in a damp and dustproof body with a high ingress protection rating, designed for use by the Army, Ministry of Emergencies and Civil Defense in conditions of significant temperature oscillations and high dustiness of atmosphere.
It can be used either independently or within the automated system of personal dosimetry control.
The dosimeter allows storing dose accumulation history with real time reference in the non-volatile memory, and communicating it to the computer. If the programmed threshold levels of gamma dose or its rate are exceeded, light and audio alarms are actuated.
DKG-21M is in operational service with the Ukrainian Army, and is successfully used by the Army and the units of the Ministry of Emergencies of Kazakhstan.
Purpose of Use:
Measurement of gamma and X-ray radiation individual dose equivalent rate (DER).
Measurement of gamma and X-ray radiation individual dose equivalent (DE).
Clock, alarm clock.
IP54 ingress protection rating.
Stand-alone use or use within automated system of personal dosimetry control.
Storage of dose accumulation history in the nonvolatile memory with real time reference.
Transfer of dose accumulation history through the infrared port to the computer.
Blocking the mode of power supply switch off until the data reading procedure finished.
Gamma radiation DER and DE threshold levels programming with the help of the computer or manually with control keys.
Blocking certain indication modes in response to the computer command.
Light and audio alarms when programmed threshold levels exceeded on DER and DE of gamma radiation.
Display automatic switch off if current gamma background is lower than the preset threshold level with instant switching on at:
pressing any control key;
gamma background increase above the preset threshold level;
alarm clock ringing.
Periodic self-testing of batteries and detector.
Energy-compensated Geiger-Muller counter.
Measurement ranges and main relative errors:
Gamma and X-ray radiation individual dose equivalent rate HP(10)
0.1…1 000 000;
(1…10 µSv/h; ±20%;10 µSv/h…1 Sv/h; ±10%)
Gamma and X-ray radiation individual dose equivalent HP(10)
Energy range of registered gamma and X-ray radiation and energy dependence
Recording resolution of dose accumulation history in the nonvolatile memory
Time of data storage in the nonvolatilememory
not less than 10
Data exchange rate through the infrared port
Positive data exchange distance between the dosimeter and the infrared port adapter